The FLO-CORP PDFlo™ PDTX2 Two Wire Digital Display Flow Transmitter continuously displays either flow rate or total and is compatible with all PDFlo Flow Meters. Precision measurement for liquid applications, from grease to liquid polymers.
Part Number: PDTX2, PDTX2-A, PDTX2-C
Replaces Part Number: FCRT-Ex10A, FCRT-Ex10C
The PDFlo™ PDTX2 is a two-wire, meter mounted flow transmitter that is compatible with the PDFlo flow meter.
The PDTX2 is used in a wide variety of applications where ease of operation and flexibility are required. A NEMA 4X
enclosure houses a large 6-digit LCD display screen where either flow rate or total is continuously displayed in the customers selected and programmed engineering units. The selection of rate, total, reset of the total, programmable sleep mode and the real-time clock can be done via magnet – without opening the enclosure. A long life field replaceable battery provides power for a period of 4 years – depending on the units configuration.
User Friendly: A large 6-digit LCD display continuoulsy indicates flow rate or total in customer’s programmed engineering units.
Easily Configurable: HART compatible with 2-wire signal so it is easily configurable from any standard HART device
Maximum Accuracy: Specifically calibrated for customer’s application
Maintenance-Free: Transmitter electronics completely isolated from process
Warranty: One-year warranty available
Power: Internal 3.6V “C” Cell Battery
Sensor/Transmitter Frequency Range: 0-3,000 Hz Max
Output Options: 4-20 mA, loop powered; Pulse output scaleable to 100 Hz
Enclosure Certifications: Class I, Groups B, C, D; Class II, Groups E, F, G; Class II ATEX: EExd IIC, IP68, NEMA 4X, FM Class No. 3615, CSA Class 1, Div 1Certification Pending
PDSG1 Positive Displacement Spur Gear Flow Meter
Aerospace, Chemical & Petrochemical, Environmental Water & Wastewater, Government Agencies, Iron, Steel & Metal, Life Sciences, Oil & Gas, Original Equipment Manufacturers, Power & Energy, Refined Products
PDTX2, PDTX2-A, PDTX2-C
|Replaces Part Number||
The CALFLO™ CFSG3 flow and temperature thermal mass flow switch is a highly precise, compact industrial instrument. The CFSG3 works according to the calorimetric principle which is evaluated by using microprocessor technology.
Driver License Terms The software you can download from this web site ("the Software") is provided by Future Technology Devices International Limited ("FTDI") subject to the license terms set out at http://www.ftdichip.com/FTDriverLicenceTerms.htm ("the License Terms"). You must read the License Terms before downloading the Software. By installing or using the Software you agree to the License Terms. If you do not agree to the License Terms then do not download or use the Software. Without prejudice to the License Terms, here is a summary of some of the key terms of the License Terms (and in the event of any conflict between this summary and the License Terms then the text of the License Terms will prevail). The license only allows use of the Software with, and the Software will only work with Genuine FTDI Components (as defined in the License Terms). Use of the Software as a driver for a component that is not a Genuine FTDI Component MAY IRRETRIEVABLY DAMAGE THAT COMPONENT. It is your responsibility to make sure that all chips you use the Software as a driver for are Genuine FTDI Components. If in doubt then contact FTDI. The Software is provided "as is''. There are no warranties (or similar) in relation to the quality of the Software. There are exclusions of FTDI liability for certain types of loss such as: special loss or damage; incidental loss or damage; indirect or consequential loss or damage; loss of income; loss of business; loss of profits; loss of revenue; loss of contracts; business interruption; loss of the use of money or anticipated savings; loss of information; loss of opportunity; loss of goodwill or reputation; and/or loss of, damage to or corruption of data. There is a monetary cap on FTDI's liability. If a custom vendor ID and/or product ID or description string are used, it is the responsibility of the product manufacturer to maintain any changes and subsequent WHQL re-certification as a result of making these changes.